Single phase peracetic acid solution. The solution consists only of 5% peracetic acid, supplemented with stabilisers and anticorrosion agents to be diluted in fresh water with a 1.16 ratio as to get a solution whose peracetic acid titre accounts for 1100 ppm.
Once the 5% peracetic acid diluted in water, the disinfecting solution is ready to be used and it should be used with 2 hours not to change its antiseptic features.
The product can be used to disinfect and decontaminated surgical instruments and probes in machines which are not using the disinfecting solution for repeated cycles.
PerActionTM One Shot is available in a 5 litre canister.
Top quality disinfection of any DM labelled device or instrument for a manual application or to be used in an instrument or probe washing machine.
Thanks to its different packaging, it complies with many different functions and requirements of clinics, surgeries and sterilisation centres.
To disinfect dental surgical tools, manually washed or in automated instrument washing machines.
Operating instructions of automated endoscope washing machines
Add the 5 litre canister content to the endoscope washing machine.
Program the machine to directly draw 28.6 ml of peracetic acid for every litre of water.
The solution antiseptic activity is active within 15 minutes.
The canister content should be entirely used 8 days after being opened.
Operating instructions for the manual disinfection
Dilute the PerAction™ One Shot solution with a 1:36 ratio (1 litre disinfectant in 35 litres of water).
5 litres of PerAction™ One Shot provide for a total 180 litre disinfecting solution.
The solution is ready to use: not waiting time for its activation and it is active for two hours.
After washing the instruments, completely immerse them in PerAction™ One Shot for 15 minutes.
Then remove the instruments, according to an antiseptic mode and rinse with sterile water.